

MoleMax 3
3 “LIVE” INTEGRATED CAMERA -
Dermoscopy system 

MoleMax 3 is the logical technology step towards an advanced Digital Image Capture System.

 

It is used for both daily ambulatory operations and for scientific research.

 

Uses patented light polarization technique without immersion fluids and /or conventional
liquid adapters.



With three particularly developed cameras for the different ELM and macro imaging requirements, it consolidates the best ergonomics and easy to use convenience.



3 Camera System For Detailed Examinations









CLINICAL MACRO VIDEO CAMERA:


	Build in Camera LEDs
	 Auto focus
	 Real Time Zooming
	 From Overview to Clinical Images
	 Polarized and Liquid Immersion Adapters
	 for ELM available
	 Variable Magnification from x20 to x100


DEDICATED ELM MICRO CAMERA:  

	 Polarized ELM Illumination
	 Fixed Focus Technique
	 Adjustable Light Intensity
	 Reproducible Skin Depth Penetration
	 Match able Color Images
	 Preset Calibration


 DIGITAL CAMERA: 

	 High resolution camera
	 Online image via USB
	 Macro und micro imaging capability
	 New polarized handle (camera control&rechargeable accu)
	Unique new camera software control
	 Camera live image on monitor





Software




REALTIME/OVERLAY FOLLOW UP
The ability to quickly compare large scale images is a key feature of  any quality skin imaging software. A  follow up image can be compared  with a historical one either side by  side or as an overlay facilitating  a simple and reliable comparison of  lesion alterations over an  extended period of time.



EXPERTIZER PLUS
The basis of this unique, future  oriented program is a   histo-pathological secure database of images of  pigmented lesions taken   with MoleMax. The  program package  includes image analysis functions  based on state of the  art technology,  and uses the well known ‘ABCD  rule’ or ‘Seven Point  Checklist’ to  support the classification of  lesions being diagnosed.



HISTOPATHOLOGY
The Histopathology module is primarily used to import Histo images into  the patient database. Histo images may be imported from any storage  media connected to the MoleMax system including the local drive, CD/DVD  drive, a USB or flash drive, external storage drive or a network  location.



TRENDING & MONITORING
Clinical Audit and Quality assessment. Macro, Micro and Histopathology  images taken over time with comments. A complete patient history can be  kept in chronological filing of every image taken with the appropriated  comments.



AUTOMATIC MOLE SCORE
To be used for automatic scoring of  moles and pigmented skin  lesions, not a final diagnosis but a given risk factor for second  opinion.



AUTOMATIC MOLE COUNT
Two similar Follow-Up Images of a relevant body site for one patient  will be compared. This software automatically detects nevi in the first  and in the follow-up image and computes the mole mapping. Additionally  the software extracts features relevant to the size, shape and  brightness of each lesion, and these features are compared for all  mapped nevi. Extracted moles are shown in full resolution giving a  detailed image of the nevus.



 
Specifications



Basic Unit
 

	Power supply and camera connections with integrated camera holder
	 Camera control electronics; incident light generator (only with MoleMax II / 3 systems)
	 Automatic mouse switch and camera recognition board (only with MoleMax II / 3 systems)
	 Latest Intel processor
	 Latest HDD for data storage
	 Latest DVD/RW for data backup
	 3.5“ Disk drive with card reader
	 Windows XP (+ latest service pack)
	 High quality video processor board and frame grabber board
	 To improve system performance, specifications are subject to change without notice.


MoleMax Macro Camera 

	 CCD PAL Color camera
	 1/3-inch sensor, 9-lens element
	 Scanning system: 625 lines, 2:1 Interlace
	 12x motor-driven zoom with software and key controls
	 Auto Focus
	 ELM Adapter for magnifications up to 100x
	 Shortest distance to subject 10mm, wide end 900 mm
	 Min. Illumination 6 lux F 1.4
	 Cable with single plug connection, 270 cm
	 Built in mouse control device (Microsoft-track stick and mouse buttons)
	 Close-up illumination feature with SMD LED light
	 Movement On-Off Sensor and automatical ELM/Macro mode recognition




 Micro Camera
 

	CCD PAL Color camera
	 1/3-inch image sensor
	 Scanning system: 625 lines, 2:1 Interlace
	 S/w ratio: 46 dB p-p/rms
	 Frequency: 15625/50 Hz
	 4-lens element, special glass x 30, fix focus
	 Cable with single plug connection, 200 cm
	 Illumination ring light and light polarisation device
	 Built in mouse control device (Microsoft-track stick and mouse buttons)


MoleMax Digital Still Camera 

	 Canon Digital Still Camera with 10 or 12 Megapixel resolution and 4-times optical zoom
	 System cable for online control of the camera
	 ELM attachment for obtaining high resolution ELM images








General Technical Data 

 The MoleMax system may be  connected to any properly grounded outlet,  which is secured with a 16A  fuse (except of explosive dangerous areas).






Nominal   Voltage: 

Power   Consumption:

Dimensions:

Weight:
110/230V   AC, 50 Hz, 1,9/1,1A 

max. 850 W

85   x 60 x 130 cm (W/L/H)

ca. 90 kg


